This 808nm hair removal handpiece model U is made of 6 bars laser stack with 300W optic power. The laser bars in this handpiece is similar with VS-MA-50-5-GTH. Differences: both positive and negative electrodes are on the top of the back side.
Most of this handpiece will be broken after 1-2millions shots. then the laser stack which is in it will burt out.
We remove the old burnt bar and replace with new bar. After that, the handpeice will become as a new one.
|>Chioce A: Send the old broken handpiece to our facatory. We will do the process of checking , repairing, cleanning and etc. Then send the fresh new handpiece back to you.||>Save Money. Replace the broken diode stack only. Low cost and convenient|
|>Chioce B: if you have basic assambly capacity, you can send the diode laser stack only to our factory. We remove the old bar and mount the new bar to recover the laser power of the stack.||>Long life time. we use diode laser bars from Germany. The newest Ausn mounting technology will be applied when repairing. After repair. The handpiece will be as a new one. life time is much longer than before|
|>Choice C: Measure your stack and comfrim the water pipe hole and screw hole position. Then press the button below to buy a ready-made stack which is match your old hadnpiece.||>All technical support available. We major in diode laser mounting, software, spare parts over 16 years. All the technology are open to you.|
|Model No.||808nm hair removal handpiece model U repair only|
|Application||808nm diode laser hair removal machine which is made in china|
|Optic Power per handpiece||300W|
|Optic Power per submounts||50W|
|Number of submounts in one stack||6bars|
|Central wavelength at 25℃||808 ± 5nm|
|Typical Operation current||40A|
|Maximum Operating current||50A|
|Typical Threshold current||7A|
|Maximum Operating Voltage||<10V|
|Fast Axis Divergence (FWHM)||39º|
|Slow Axis Divergence (FWHM)||8º|
|Operation Condition||non-condemsing atmosphere, cleaning room|
|Expected lifetime||10millions shots|
|Typical pitch per bar||2mm|
|Flow rate||3L/min +-10%|
|Water Quality||pure water, mixed bed ion exchanger,particle filter<25µm|
|cooling system||Do not use any material that combination|
More question, please click here to email me.